Subject Interface Stimulation
The Subject Interface Module (SIM) is the control object for the SI2, SI4, and SI8 devices. It connects to a special DSP-M card in your RZ processor. The Subject Interface is configurable with stimulator cards (IZV), analog amplifier cards (PZA), or digital headstage interface cards (PZD). Because of the diversity in functionality of this device, each card type has its own "Sub-HAL" object within Synapse that is used to configure those cards. The acquisition and stimulation all happen within the Sub-HALs, which are all independent objects in the Processing Tree.
See the Electrical Stimulation Guide for help with experiment design.
The IZV can split the available boards into sub-stimulators which are completely isolated from one another. It is typically controlled by the Electrical Stim Driver gizmo.
|IZV Block Diagram|
|IZV Stimulator Configure Tab Options|
When using Sub Stimulators, each sub-stimulator must be configured independently. Each sub-stimulator has its own multi-channel input signal. For example, the first sub-stimulator uses the gizmo input called "VoiceIn-A", the second sub-stimulator uses "VoiceIn-B" gizmo input, and so on. It is generally expected that this signal comes from an Electrical Stim Driver gizmo so that it has the proper format. The format is a multi-channel signal where the odd channels contain the actual stim waveforms and the even channels contain the target stim channel. For an input that has 4 unique stim signals, it will be an 8-channel signal where channels 1, 3, 5, 7 are the stim signals and channels 2, 4, 6, 8 are the corresponding target stim channels.
Each physical IZV board has 4 unique hardware voices that can be routed to its 16 output channels.
The data table at the bottom shows the mapping between the gizmo input voices and the physical hardware voices. The first column is the voice of the "VoiceIn" for this sub-stimulator. The middle column is the board number within the sub- stimulator that it uses and the hardware voice on that board it uses (a, b, c, d). The last column is the range of channel numbers that the VoiceIn stim channels should address.
|IZV Stimulator Compliance and Safety Tab Options|
See the SIM section of the System 3 Manual for more information about configuring and using the IZV interface.
|States of the IZV Runtime Interface|
If Safety Mode is enabled on the Safety and Compliance Tab, then an LED will show the state of the compliance monitor. If Remote Arming is disabled, the Arm/Disarm button is shown and the device can be armed when ready.
The system can only Arm if the fiber is connected and a zero current command is sent to the SIM device, so make sure the gizmo controlling stimulation (typically an Electrical Stim Driver) is muted.